Labview Error Code 21
To correct this error, install the latest version of DirectX from http://www.microsoft.com/downloads. 4810 Cannot recognize sound format. My Profile | RSS | Privacy | Legal | Contact NI © 2014 National Instruments Corporation. The disk or hard drive does not have enough free space to complete the operation. The target version for save must be a valid LabVIEW version in the form x.x or xx.x, where x is a number. Check This Out
For example, this error might occur when a front panel terminal is inside a structure. 1430 The path is empty or relative. You do not have the correct permissions for the file. All rights reserved. | Cart|Help KnowledgeBase Request Supportfrom an engineer NIHome > Support > KnowledgeBase English 13 ratings: 3.61 out of 5   Error 21 Void can be used only as return type of function. 1186 Cannot show or hide the label on its own. http://digital.ni.com/public.nsf/allkb/78579B0FAA17F92286256993006A6473
Due to race conditions that can occur when you have a Boolean value with latching mechanical action, you cannot programmatically read Boolean values that are set with a latching mechanical action. A buffer underflow has occurred because the application is not writing data quickly enough. −4803 The sound driver or card does not support the specified operation. The sound driver or card cannot accommodate the specified configuration. Save the new VI or function to a different directory than vi.lib so you do not overwrite the original. 44 Object ID too low. 45 Object ID too high. 46 Object
Answered Your Question? In the National Instrument Installers to Include list box, select the appropriate installers. Build the installer and install the application on another machine. The file might have been moved or deleted, or the file path might be incorrectly formatted for the operating system.
A waveform must have 0 or 1 dimensions. 1177 Invalid data type for Call Library Function Node parameter. The Connectivity VIs are available for free download from ni.com. To correct this error, ensure that the names you use for drag data are unique in each element of the data array for the method or event. 1385 Cannot start a http://digital.ni.com/public.nsf/allkb/9DB0FFEA194AA7D086256C10007C8023 My Profile | RSS | Privacy | Legal | Contact NI © 2014 National Instruments Corporation.
Try freeing up disk space or saving to a different disk or drive. 10 Duplicate path. 11 Too many files open. 12 Some system capacity necessary for operation is not enabled. You also can specify the path of these DLLs with the set path command in the autoexec.bat file or list it in the path environment variable on Windows. There are a few possible causes for this depending on what you are doing in your code. This file cannot be saved until all dependent files have been named. 1022 Wizard Template not found. 1023 Wizard template does not have a diagram. 1024 Call Instrument aborted. 1025 Application
This could be caused by an invalid sound driver. 4801 Invalid sound task refnum. look at this web-site when I try running my program on a different machine I get the following ERROR:"Missing subvi DAQmx Read.vi ... But the new system having labview6.1 and labview 7.1 run time engine with NI scope 2.6. For example, use \ as path separators on Windows, : on Mac OS X, and / on Linux.
is FALSE and a queue or notifier with that name was not found, LabVIEW returns this error. 1102 The string wired to the xml string input is empty. 1103 The XML his comment is here Your post has nothing to with the thread you posted into. Verify that the VI is marked to save its compiled code separately and that it is not broken. Please give me an solution..
LabVIEW Register for the community · Log in · Help CommunityCategoryBoardDocumentsUsers turn on suggestions Auto-suggest helps you quickly narrow down your search Please tell us why. There are duplicated items in the array. 1304 The array index is outside of the array bounds. 1305 The required page cannot be found. 1306 Unable to load new code resource http://distributedsimulation.com/labview-error/labview-error-code-91.html Primary Software: NI Installers>>Deployment Framework Primary Software Version: 1.0 Primary Software Fixed Version: N/A Secondary Software: N/A Problem: When building an installer using LabVIEW Application Builder, the LabWindows/CVI Build Distribution feature,
Poor|Excellent Yes No Document Quality? When you obtain a reference to a queue or notifier, you can specify which data type this mechanism stores. The palette type value has enumeration values of Controls or Functions.
The return type must be Void, Numeric, or String.
No reference could be returned. Recompiling the VI may fix this issue. 4800 Selected Device is Invalid The currently selected device index is invalid. The input for class name is not correct or is in the wrong format. Make sure you include the front panel of the VI when building a stand-alone application. 1014 Linker error. 1015 Printer is not responding.
There is already a VI in this hierarchy with this name. 1449 Arrays must have at least one dimension and a maximum of 63 dimensions. 1450 One or more untitled library Aida Member 06-28-2006 05:42 AM Options Mark as New Bookmark Subscribe Subscribe to RSS Feed Highlight Print Email to a Friend Report to a Moderator Hi Sarah, thanks for your reply...I Select File»Open to open the VI and then verify that you are able to run it. 1004 The VI is not in memory. 1005 VI execution has been disabled in the navigate here Poor|Excellent Yes No Document Quality?
LabVIEW determines how data is aligned and you cannot alter that alignment. See Related Links for NI-DAQmx Driver Page If you are doing GPIB communication, serial communication, or any other protocol that uses NI-VISA, make sure you have installed the NI-VISA Run Time The specified wave format is not supported. You must load an instance of the polymorphic VI instead of the polymorphic VI itself. 1125 File version is later than the current LabVIEW version. 1126 VI version is too early
The operation is not allowed on this kind of control or a control at this level. 1087 There is no DataSocket information available for the object. 1088 Bad value for parameter. This error might occur if you remove the front panel of the VI when building a stand-alone application. 16 Image not found. 17 Not enough memory to manipulate image. 18 Pen For example, the Application:All VIs In Memory property is allowed locally, but remotely you should use the Application:Exported VIs In Memory property instead. RavensFan Knight of NI 01-04-2012 08:55 PM Options Mark as New Bookmark Subscribe Subscribe to RSS Feed Highlight Print Email to a Friend Report to a Moderator Translation: I have
My Profile | RSS | Privacy | Legal | Contact NI © 2014 National Instruments Corporation. This error can occur for several reasons. Verify the values you wired to the Call Library Function Node. 1098 Cannot disconnect type definitions or polymorphic VIs if the block diagram is not available. 1100 No object of that Eso tiene tambien la instalacion de Labview 7.0...
All rights reserved. | Cart|Help KnowledgeBase Request Supportfrom an engineer NIHome > Support > KnowledgeBase English 3 rating: 1 out of 5   LabVIEW Load
© Copyright 2017 distributedsimulation.com. All rights reserved.